This calibration manual focuses on performance verification and calibration of the Fluke. A and A Digital Multimeters (hereafter referred to as the. The Fluke A & A digit precision multimeters have the accuracy & versatility to handle your most demanding measurements, on the bench or in a. Manual/User Manual on CD-ROM, FVF-BASIC. FlukeView Forms Software Basic Version. Ordering Information. Fluke A. Digit Precision. Multimeter.
|Country:||Bosnia & Herzegovina|
|Published (Last):||3 March 2004|
|PDF File Size:||11.9 Mb|
|ePub File Size:||7.36 Mb|
|Price:||Free* [*Free Regsitration Required]|
Also increases or decreases the range when in manual ranging mode.
The functions of a counter, capacitance meter, manuall thermometer are built in for unparalleled versatility. Fluke A Manual pages. Instrument programming drivers, DLL files and support information are included to get your test system up and running as quickly as possible.
Previously laid down plot marks are adjusted down in size proportionally to the new vertical axis.
Fluke 8845A User Manual
Chapters Table Of Contents If you want to store the present Meter configuration in a location other than the first four locations, press the MORE soft key.
The Fluke A and A 6. All measurements use the Input LO connector as a common input. Petrochemical industry calibration solutions 2. Flke Double insulated Do not dispose of this product as unsorted Static awareness.
A 3-pole analog filter for mamual noise immunity. Accessories Table lists the available accessories for the A and A. To make a capacitance measurement: Summary of Contents for Fluke A Page 1: Majual 4-wire measurement with only two leads Patented split terminal jacks for the 2 x 4 ohms function allow you to perform 4-wire measurements using only two leads instead of four. The manual is divided into the following chapters: Selects between manual and auto range mode.
Page of Go.
8845A/8846A 6.5 Digit Precision Multimeters
It contains all of the information a new user will need to operate the Meter effectively. Press Lto see the measurement setup menu. RS Cable RS43, 2 m. Testing Continuity Continuity testing determines whether a circuit is intact i.
Power and utilities industry calibration solutions 5. The filter is A FLTR active when this soft key label is highlighted and will increase stabilization time of the measurement.
A/A Users Manual
Input connectors source fulke for 4-wire Ohms measurements. If the number of samples is set to a value greater thanconsecutive memory files will be used to store all mnual samples. Transfer data points from your meter to your PC and use the two standard, noncustomizable forms to display your readings in table or graphical form. Memory enabled for storing readings. Extinguishes when last sample is stored. Carefully unpack the Meter from its shipping container and inspect the contents for damaged or missing items.
After removing the power-up configuration, power the Meter down and back up using the rear-panel power switch to restore factory default settings. This chapter explains the function and use of the controls and indicators located on the front and rear panels of the Meter. Read “Maintaining in-situ traceability on the factory floor with process metrology.
Press S on the A or Q on the A.
FLUKE A USER MANUAL Pdf Download.
Enter text from picture: Page 95 Appendices Errors Incorrect packet size from inguard Info packet rec’d; link not active Inguard Calibration Constant majual failed Inguard not responding recv Inguard not responding send INITiate received but was ignored Instrument configuration load failed Instrument configuration store failed Insufficient memory Invalid dimensions in a channel list Invalid parameter Invalid parameter Use only Fluke fuses A good example of this is using the filter when measuring the DC value of a signal with AC content, such as a DC power supply voltage that has significant AC line ripple present.
To manually stop the measurement collection, press the STOP soft key from the statistics menu.
Each offers a full range of support services. Using the Analyze Functions The Meter is capable of performing mathematical operations with measured values, as well as tracking a series of measurements. Log live readings while connected to a PC or store, display and share readings in interactive forms, graphs and tables. Checking Diodes To check a diode: First verify that the line voltage selection is set to the correct 8846s, and then that the correct fuse for that line voltage is installed.
Are you looking to establish traceable calibration of factory test systems on the factory floor? To reveal signal quality issues like drift, intermittants and stability, view data as a real time trend plot or histogram, with the unique analyze mode. USB port on the front panel of the Meter.
If the fuse is blown, the Meter will read overload. Appendices Errors Incorrect packet size from inguard Info packet rec’d; link not active Inguard Calibration Constant write failed Inguard not responding recv Inguard not responding send INITiate received but was ignored Instrument configuration load failed Instrument configuration store failed Insufficient memory Invalid dimensions in a channel list Invalid parameter Invalid parameter All product names are trademarks of their respective companies.
You can also use the A and A to measure frequency and period. XW Warning To avoid electric shock or fire, do not use makeshift fuses or short-circuit the fuse holder.